- BrandБаромембранная Технология ООО
- Country of manufactureRussia
The device of vacuum filtering provides fast filtering of test and plums of the filtered water in the sewerage. It is made under the membrane Ø142 filter of mm.
Device complete set:
- filtering cell
- vacuum station
- intaking and filtering device
- set of connecting hoses.
Characteristics of components of the PVF-142 device B
1. Filtering cell
for filtering of water under vacuum. Consists of a cover and the basis at which top end face titanic frit for placement of the disk filter is established. The cover and the basis are fixed among themselves by means of a clip. Has three basic legs.
|diameter of the membrane filter||142 mm|
|overall dimensions||180 mm x 210 mm x 310 mm|
2. Vacuum station (patent of the Russian Federation RU (11) 2309785 (13) C2)
for creation of vacuum with use of pressure created by the built-in pump. Includes the pump on a frame, the ejector and buffer capacity. The station is supplied with cranes and the vacuum gage.
|working vacuum||0,35-0,96 kgfs/cm2|
|expense on water||3,5-5,0 l/min|
|power consumption||0,6 kW|
|power supply network||single-phase, 220B, 50 Hz|
|overall dimensions||400 mm x 280 mm x 140 mm|
fixes a hose at the bottom of the container with the studied water. Represents the union with the flat basis at which lower end face there are four radial grooves for water pass.
|overall dimensions||48 mm x 28 mm|
4. Intaking and filtering device
fixes a hose at the bottom of capacity or in an open reservoir and it is applied when sampling water with the high content of the weighed substances. Represents the union with the welded two flat disks. On the lower disk the punched leaf on which the fabric filter is fixed is fixed. On the top disk three curved support are fixed.
|overall dimensions||180 mm h180 mm x 60 mm|
5. Set of connecting hoses
- the hose is vacuum silicone
- hose pressure head PVC
- PVC tube