System of out-patient registration of the ECG Web Holter Beecardia
The registrar Bee-S is the modern tiny registrar of the ECG of professional quality allowing to make registration, the analysis and interpretation of electrocardiograms as by means of the application on the personal computer, so on the Internet through Android of the tablet, the smartphone, the laptop or the personal computer.
Execution options for registrar Bee-S:
5 - 3/7/12-channel;;;;;;;;;;;
3 - 2-channel...........
The system of daily monitoring of the ECG on Holtera (System) is intended for long-term continuous registration of the ECG on three assignments from the subsequent processing and the analysis of the received records. The system allows to reveal a wide range of violations of work of heart, to define and predict emergence of ischemic states at surveyed, to edit results of the automatic analysis in the dialogue mode, to form the text conclusion including illustrative graphic material (typical fragments of the ECG, graphics of derivative parameters) at the choice of the user. Provides automatic detection of frequency of occurrence of separate types of violations of a rhythm throughout the studied period.
The registrar Bee-S can be used as Holter the monitor.
Electrocardiographs are intended for application in offices of functional diagnostics of policlinics, clinics, hospitals, offices of cardioresuscitation and intensive therapy of cardiological offices of hospitals, at visit of patients by ambulance crews and district doctors at home, and also for independent removal of electrocardiograms by patients.
Works with the mobile device or the personal computer through Bluetooth. Turns Android - the smartphone or the tablet into the professional portable electrocardiograph.
Uniform software compatible to Windows, Mac OSX, Linux and Android. Work with Apple iPad is possible under an operating condition of the program on the device from one of the above-stated operating systems. The software is integrated with the Beecardia Cloud Platform.
The software allows:
to remove and write down a qualitative ECG signal of rest;
to filter a signal the best algorithms that gives the chance to restore the ECG even from strongly noisy signal;
to form the conclusion;
to load researches into a cloud service;
to synchronize lists of patients, records with all devices.
Web service allows:
To observe patients in any place, at any time
To simplify communication between cardiologists
to provide to patients access to their researches.
to watch dynamics of work of heart
to be examined by professional cardiologists without leaving the house.
to request the second opinion from other doctor
Gives the chance for training of students of medical schools with use of the ECG largest library of signals
Registrars can work in two modes:
In this mode the registrar synchronously removes 3 standard assignments: I, II, Vx (any of chest). With the subsequent standard (long ago accepted in world practice) reconstruction of seven assignments of the ECG: I, II, III, aVR, aVL, aVF and one of chest Vx.
Arrangement of electrodes:
Red: in the right subclavial area is closer to a shoulder;
Orange: symmetrically red in the left subclavial area is closer to a shoulder;
Green: at the left over an inguinal sheaf;
White: the axillary line left so-so at the level of the 5th mezhreberye (V6)
or the sredneklyuchichny line at the level of the 5th mezhreberye (V4), or a position of any chest assignment;
Black: the middle of a breast (a neutral electrode it is possible to fix in any place,
which is less subject to physical activity).
EASI mode - Orthogonal system of assignments X, Y, Z.
Assignment of X and assignment of Y correspond to assignments of AI and ES and in accuracy display a condition of the ECG of a vector in the frontal and horizontal plane respectively. Assignment AS (Z) not only sagittalny projection, but also a little frontal.
Arrangement of electrodes:
White (E): along a breast between the electrodes marked as "A" and "I" it is horizontal with the level of the 5th mezhreberye;
Red (And): the left middle line at the level of the 5th mezhreberye;
Green (S): it is located along the top breast below its corner;
Orange (I): the left middle line at the level of the 5th mezhreberye;
Black (N): a neutral electrode it is possible to fix in any place which is less subject to physical activity.
According to repeatedly checked algorithm, by means of imposing in the EASI system it is possible to restore the standard 12-channel ECG.
ECG cable 3 or 5 of wires
ECGs channels of 2 channels (I, II, III, aVR, aVL, aVF, Vx) or
3/7/12 channels (I, II, III, aVR, aVL, aVF, V1-V6)
Type of the Button ECG cable
Food Built-in accumulator
Android communication phone/tablet or personal computer
Frequency range is 0.05 - 150 Hz
Filters Surge protector of 50 Hz; isoline filter
Word length of ATsP of 24 bits
Frequency of sampling is 4 00 Hz
Weight of 40 g
Main functions of the program of the analysis:
Control of process of input of an electrocardiosignal before the beginning and at any moment of inspection
Automatic calculation of parameters of the daily ECG, with classification of events and automatic detection of the QRS types of complexes
Full automatic or semi-automatic (with participation of the doctor in the dialogue mode) the analysis of records ECGs with issue of the conclusion on main types of violations of a rhythm and conductivity
Assessment of statistical characteristics of arrhythmias with formation of "rhythm profiles", registration of change of amplitude of a segment of ST on each of assignments, detection of episodes of ischemic changes of the ECG throughout inspection
Assessment of variability of a warm rhythm
Web service allows:
To observe patients at distance
To read the ECG and clinical records where you were - via the computer or mobile
To request or provide the second opinion on the Internet
To provide to patients access to researches on the Internet for receiving