The automated complex for graduation of sensors of inclinometers
- Customer pickup,
- In detail
- Country of manufactureRussia
TECHNICAL AND METROLOGICAL CHARACTERISTICS
THE AUTOMATED COMPLEX FOR TARIROVANIYA OF SENSORS
INCLINOMETERS LIKE KSVSH.611156.003 "ION"
The complex of JSC TDI allows to make in the automatic mode periodic (time in 4-6 months) the tarirovaniye of sensors of inclinometers of ION-1, ION-2 and ION-3 consisting in measurement of errors of sensors and formation of the new correction file necessary for reduction of errors of inclinometers on an azimuth and an antiaircraft corner to admissible values.
Use of JSC TDI allows to reduce by 2,5 - 3 times time for carrying out operation of a tarirovaniye of sensors of inclinometers like "ION" and to facilitate work of the operator in comparison with the manual workplace demanding long, laborious procedure.
The software of JSC TDI functions in an operational environment "WINDOWS - 95, 98, HR, vista, 7, 8".
Technical and metrological characteristics of JSC TDI
|Range of reproduction of tarirovochny corners, hail.
||0 - 360|
|Limit of the allowed main error of reproduction of tarirovochny corners, min., no more||±0,6|
|Outer diameter of tariruyemy blocks of sensors of inclinometers, mm||42, 60, 73|
|Length of tariruyemy blocks of sensors of inclinometers, mm, no more
|Mass of tariruyemy blocks of sensors of inclinometers, kg, no more
|Time of a tarirovaniye of one inclinometer, hour, no more||2|
|Electric food of JSC TDI tension||220 V, 50 Hz|
|Service life||6 years|
|Guarantee maintenance during||12 months|
Are included in the package of JSC TDI:
- stand of JSC TDI;
- PEVM set with the software;
- set of means for a tarirovaniye of sensors of inclinometers of ION-1;
- set of panels of means for a tarirovaniye of sensors of inclinometers of ION-2, ION-3.
Appearance and overall dimensions of the stand of JSC TDI are given in figure 1
Figure 1 - A stand design with a DB, located in the plane of a magnetic meridian.
- - rack with wheels and support;
- - arm;
- - the pipe bearing;
- - the pipe rotating;
- - shaft of rotary part;
- - a clutch for horizontal fixing of the block of sensors;
- - cam switch;
- - flood mechanism;
- - reducer;
- - fixing mechanism;
- - a clutch for fixing of the block of sensors in the vertical plane;
- - counterbalance;
- - counterbalance;
- - model sensor;
- - casing;
- - flange;
- - an arm for fastening of BE the ION;
- - a reducer with the electric motor.